## How to set the maximum flow rate of the Coriolis mass flow meter?

How to set the maximum flow rate of the Coriolis mass flow meter?

The Coriolis mass flow meter has a certain reduction the diameter in the design and manufacturing process, that is mean the sum of the cross-sectional areas of the two flow tubes is smaller than the flange cross-sectional area. Thus, the medium has varying degrees of acceleration as it flows through the flow meter. The maximum flow rate is an important parameter in the industrial control process, which has an impact on the process and safety. Therefore, some users will ask for the limit of the maximum flow rate. The flow rate of the mass flow meter when measuring the gas is often much larger than the flow rate when the liquid is metered. At high flow rates, the friction between the gas medium and the wall of the flow meter will generate a large noise, which will affect the signal of the flow meter. Quality, therefore, it is recommended that the maximum flow rate when measuring gas using the eYc FCM06 Coriolis mass flow meter should not exceed one-third of the speed of sound.