Exhaust gas treatment flow chart:
Exhaust gas scrubber uses the contact between gas and liquid, and transporting gas from the contaminants into the liquid, then separate the clean gas from the contaminated liquid to achieve clean air, and use average flow measuring tube or pitot tube collocation differential pressure transmitter to achieve flow monitoring.
Semiconductor manufacturing or electronics frequently etched or cleaned with a large amount of inorganic acid-base solution or gas enthalpy, some chemicals are collected by the system due to volatilization or emissions, the semiconductor industry emission standards should confirm one of the two requirements for removal efficiency or total emissions. For example, pollutants such as nitric acid, hydrochloric acid, phosphoric acid, and hydrofluoric acid, the total emissions of the plant must be less than 0.6 kg/hr, the sulfuric acid needs to be less than 0.1 kg/hr, or the removal efficiency of the scrubber is over 95%.
Note: Unable to prove compliance with emission standards, the control conditions should meet the following requirements: Equipment washing cycle sink pH should be greater than 7、Wetting factor should be greater than 0.1 m2 /hr、Filled section empty tower dwell time should be greater than 0.5 seconds and the specific surface area of the filler should be greater than 90m2 / m3.
Consider the characteristics of exhaust gas have the following points: handling air volume、density、temperature、pressure、corrosive、water content、smelly odor、Inflammability、explosive nature、toxicity、stickiness、Therefore, it is necessary to choose high temperature resistance、corrosion resistant、resistant dust average flow measuring tube (pitot tube), can do it correct exhaust gas flow monitoring.
eYc AFMT Average Flow Measuring Tube, operating pressure Max 10 bar, operating temperature Max. 250°C, Measuring medium / coefficient is air / flow coefficient (K) : 1, measuring tube use SUS316.
In the flow system, space problems often result in the inability to give enough straight pipe sections, causing serious turbulence problems.
AFMT can average the characteristics of the flow curve of the fluid in the pipeline.
eYc AFMT inserted into the duct(along with the whole diameter) to measure the flow. When the probe encounters the flow, will sense and get the average total pressure P1 in windward side and static pressure P2 in leeward.
AFMT then gets the flow velocity by measuring the difference between total pressure and static pressure (i.e.) dynamic pressure (∆P) and flow average velocity (V). Stainless steel housing, high temperature resistant, corrosion resistance, dust resistance, can measure micro flow rates. Usually collocation eYc PHM33 Differential Pressure Transmitter measuring flow rate and flow rate in the calculation of flow rate in the calculation, principle uses very small amounts of gas, passing it over the airflow path of the sensor body, and combined with integrated circuit to convert into differential pressure signal, and use ModBus RS-485, MODBUS RTU communication protocol set can be transferred to PLC or Industrial PC for remote monitoring, calibration correction only need to correction differential pressure transmitter, avoid disassembling the upper flange of the pipeline.
Suitable for air pollution control or poor working conditions such as petrochemical industry、electronic industry exhaust gas exclusion, environmental protection engineering installation position is not limited.
eYc AFMT Average Flow Measuring Tube ( Pitot tube ) used for exhaust gas scrubber installation diagram (below):
Another option is eYc AFMS-160 S-type Pitot tube collocation PHM33 Differential Pressure Transmitter, operating pressure max 10 bar, operating temperature 800°C, measuring medium/coefficient is air/flow coefficient(K) : 0.84, measuring tube use SUS316, designed for measuring the flow velocity of the gas fluid in the chimney and the environment, prevents clogging of soot under harsh conditions, further rapid reaction treatment of smoke and sewage discharge procedures.
S-type pitot tube for measuring the local velocity of a fixed point, can be applied in scientific research、produce、surroundings protection、mine ventilation and tunnel construction, use S-type pitot tube to measure airflow velocity in the pipeline, it to determine flow and speed by conversion, easy to use and accurate, is a widely used method for measuring gas flow pressure. high temperature, corrosion-resistant, resistant dust, large, open tip design resists fouling, can be used in harsh environments, exhaust gas emission, environmental protection engineering.
Similar product recommendation
eYc AFMT Average Flow Measuring Tube ( Pitot tube )
eYc AFMS-160 S-type Pitot tube